John Loram <johnl@...>
When you splice a connector into a thermocouple wire, you actually create
two junctions (the point I missed ): one junction is created where the
thermocouple wire is attached to the plug side of the connector, and a
second junction is create where you connect the thermocouple wire to the
socket side of the connector. These two junctions are of opposite polarity,
so, if the two junctions are at the same temperature (as they would be since
they are plugged into one another) they electrically balance out one another
and will have no effect on the temperature measurement.
building like crazy, -john-
From: Q-LIST@yahoogroups.com [mailto:Q-LIST@yahoogroups.com] On Behalf Of
Sent: Thursday, February 04, 2010 3:11 PM
Subject: [Q-LIST] Re: Thermocouple compatible bulkhead connector?
I was curious about this as in theory the additional dissimilar metal
junction will cause errors in the readings. As an experiment I tried
connecting a type K thermocouple to a digital thermometer readout with a
length of copper wire spliced in. With the thermocouple immersed in a hot
oil bath I watched the temperature displayed as I used a hot air gun to heat
the junction of copper to thermocouple wire. I thought I could see 1-2
degrees change heating the junction from ambient to as hot as the gun would
deliver, but it was difficult to tell the change from random drift.
I have forgotten if I repeated the experiment with a type J thermocouple.
The metals involved are different and I would not assume the same results
So my conclusion is, for type K as used in EGT probes and often used for
CHT, you can use the non-thermocouple connector or even use copper extension
wire without significant error. And as long as all the cylinder's wires are
the same temperature at the change of metal, all the errors are the same
magnitude and direction, so with EGTs where only the trend is important, you
are doubly ok.
--- In Q-LIST@yahoogroups. <mailto:Q-LIST%40yahoogroups.com> com, "John
Loram" <johnl@...> wrote: